﻿ Variable pathlength cell
Wed . 20 Jul 2020
TR | RU | UK | KK | BE |

# Variable pathlength cell

variable path length cellulitis, variable path length cell phone
A variable pathlength cell is a sample holder used for ultraviolet–visible spectroscopy or infrared spectroscopy that has a path length that can be varied to change the absorbance without changing the sample concentration

## Contents

• 1 Equations
• 2 Applications
• 3 Experimental methods
• 4 Background subtraction
• 6 References

## Equations

The Beer-Lambert law states that there is a logarithmic dependence between the transmission or transmissivity, T, of light through a substance and the product of the absorption coefficient of the substance, α, and the distance the light travels through the material ie the path length, ℓ The absorption coefficient can, in turn, be written as a product of either a molar absorptivity of the absorber, ε, and the concentration c of absorbing species in the material, or an absorption cross section, σ, and the number density N of absorbers see Beer Lambert Law link for full derivation

BSA linearity A = ε ℓ c

Spectroscopy with a variable pathlength cell takes advantage of Beer-Lambert law to determine concentrations of various solutions By knowing the molar absorbtivity of the material and varying the path length, absorption can be plotted as a function of path length See sample plot to the right:

By taking a linear regression of the linear plot above an expression relating Absorbance, A, slope, m, pathlength and concentration can be derived

A linear equation of two variables can be derived,

y = m x + b

by equating in terms of units we get,

A = m ℓ + b

Since the slope of the line is in units of Abs/Pathlength, slope can be expressed as,

m = A ℓ }

by inserting into Beer’s Law we get,

m = ε c

This is the slope spectroscopy equation

## Applications

Variable pathlength techniques can be applied in any situation where Beer’s law can be applied It provides an analytical method that averages out minor variations in sample preparation consistency It also provides a means to calculate concentrations without calibrations curves or serial dilution of samples

Variable pathlength absorption spectroscopy is typically used when highly reproducible data is a necessity This can be in the fields of medicine, biotechnology, pharmacology, and drug discovery It is particularly useful in the protein purification stage of biotechnology where accurate concentrations of various proteins are required or in crystallography

Determining the relative ratio of protein to DNA is common practice and can be calculated by finding the slope at the corresponding absorption peaks and taking their ratio This method is used to find the purity of a sample containing these two types of molecule

## Experimental methods

In ultraviolet-visible spectroscopy or spectroscopy in general a 1 cm pathlength cuvette is used to measure samples The cuvette is filled with sample, light is passed through the sample and intensity readings are taken The slope spectroscopy technique can be applied using the same methods as in absorption spectroscopy With the advent of accurate linear stages, variable pathlength absorption spectroscopy is easily applied experimentally

Other experimental methods include using ratios of slopes to build extinction coefficient spectra This is possible because application of slope spectroscopy allows the scientist to keep concentration levels constant and vary path lengths

## Background subtraction

Base line no base line

Variable pathlength absorption spectroscopy uses a determined slope to calculate concentration As stated above this is a product of the molar absorbtivity and the concentration Since the actual absorbance value is taken at many data points at equal intervals, background subtraction is generally unnecessary The image on the right is a linear plot showing both the background corrected data and the raw data

This shows that the absorbance values on the plot are offset by an equal amount and the slope of the two plots are equal Thus, the concentration calculated from the two plots is equal Other scalar components that contribute to the absorbance of a given sample like contaminants on the cuvette or a different cuvette material also are averaged out during the slope measurement

The technique is also applicable for in line measurements for TFF and chromatography applications[citation needed]

• Applied spectroscopy

## References

1. ^ Teresa Nowicka-Jankowska December 1986 Analytical visible and ultraviolet spectrometry Elsevier p 124 ISBN 978-0-444-42371-9
2. ^ Doucen, R Le; Houdeau, J P; Cousin, C; Menoux, V 1985 "Variable path-length, low-temperature cells for absorption spectroscopy" Journal of Physics E: Scientific Instruments 18 3: 199–200 Bibcode:1985JPhE18199L doi:101088/0022-3735/18/3/007 ISSN 0022-3735
3. ^ Stewart, Sarah A; Sommer, André J 1999 "Variable Path-Length Cells for Discovery-Based Investigation of the Beer-Lambert Law" Journal of Chemical Education 76 3: 399 Bibcode:1999JChEd76399S doi:101021/ed076p399 ISSN 0021-9584
4. ^ Flowers, Paul A; Callender, Sherry-Ann 1996 "Variable Path Length Transmittance Cell for Ultraviolet, Visible, and Infrared Spectroscopy and Spectroelectrochemistry" Analytical Chemistry 68 1: 199–202 doi:101021/ac950580w ISSN 0003-2700 PMID 21619236

• Thakkar, Santosh V; Allegre, Kevin M; Joshi, Sangeeta B; Volkin, David B; Middaugh, C Russell 2012 "An application of ultraviolet spectroscopy to study interactions in proteins solutions at high concentrations" Journal of Pharmaceutical Sciences 101 9: 3051–3061 doi:101002/jps23188 ISSN 0022-3549
• Scott Huffman, Keyur Soni and Joe Ferraiolo UV-Vis Based Determination of Protein Concentration: Validating and Implementing Slope Measurements Using Variable Pathlength Technology by September 2014 http://wwwbioprocessintlcom/manufacturing/antibody-non-antibody/uv-vis-based-determination-protein-concentration-validating-implementing-slope-measurements-using-variable-pathlength-technology/

Spectroscopy at Curlie based on DMOZ

variable path length cell, variable path length cell membrane, variable path length cell phone, variable path length cell phones, variable path length cello, variable path length cellular, variable path length cellulite, variable path length cellulitis

## Variable pathlength cell Information about • Variable pathlength cell beatiful post thanks!

29.10.2014

Variable pathlength cell
Variable pathlength cell
Variable pathlength cell viewing the topic.

### B♭ (musical note)

B♭ B-flat; also called si bémol is the eleventh step of the Western chromatic scale starting from C ...

### Fourth dimension in art

New possibilities opened up by the concept of four-dimensional space and difficulties involved in tr...

### Holt Renfrew

Holt, Renfrew & Co, Limited, commonly known as Holt Renfrew or Holt's,1 is a chain of high-end C...

### Later Silla

Later Silla 668–935, Hangul: 후신라; Hanja: 後新羅; RR: Hushila, Korean pronunciation: ...